|Contact Us For Quotation
Xtester-ZX99E,ZX95E- Electric Decade Resistor Variable Resistance Box Experimental Equipment For DC Desistor
|Type ZX99E DC resistor is a multiple - valued resistor, formed by decade resistor groups in series. The resistance between both fasteners can be changed with the position of the switch brush contact.
The resistor is miniature and portable. The switches are fully enclosed, no .need to clean, low pressure of contact , long life time and less variations. The stand of the resistor is made of plastic so that the insulation resistance is high. The readings of resistance on the panel is lined up, and can be read forthright and clearly. The resistance elements are all wound noninductively with enameled manganin wire, and go through strict technical process, in this way, the resistor has high stability and can be used in any position.
1. It is suitable for electric measurement and other experiments in factories, schools and scientific research units.
2. It is used as adjustable precision resistor in DC circuit.
3. It is even more suitable for calebration of resistance thermometer.
Reference and Nominal Temperatures:
REFERENCE RANGE NOMINAL RANGE 20+1.5 °C 20±15°C
Reference and Nominal Relative Humidity Range:
REFERENCE RANGE RELATIVE RANGE 40% -60% 25% -80%
Under the conditions of reference temperature and power, relative humidity is beyond reference range, but within nominal one, an additional error due to relative humidity should be within relevant class index.
Under the conditions of reference relative humidity and power, temperature is beyond reference range,but within nominal one, an additional error due to temperature should be within relevant class index.
Under the conditions of temperature, relative humidity applied power is beyond reference range, but within nominal one, and an additional error due to power should be within relevant class index.
The resistor can be acted on by ultimate power for 2 hours. The difference between the value of the resistor before and after the application of this limiting power dissipation shall not exceed 10% of a value corresponding to the class index when the resistor is again under reference conditions.
The insulation resistance between circuit of the resistor and metal chassis should be over 100MQ.
A 45~65Hz, 2000V AC sinusoidal voltage should be beared for 1 minute of test between the resistor circuit and its cover without breaking down.
Dimensions and Weight:
TYPE DIMENSIONS(mm*mm*mm) WEIGHT(kg) ZX99 327X95X94 1.35